Vakumat 500 Dental Vacuum Furnace
I obtainted a non-working Vakumat 500 dental vacuum furnace, and thought it would be fun to see what it takes to get it working. The control panel is missing, but it's connected via a DB9 connector - so my guess was that it probably uses a serial line. Maybe I can just extract the firmware, look at what the commands look like and make a replacement control panel?
1: Opening and Mechanical
The furnace itself came with some mechanical damage from shipping. I bent it back in shape at best as I could after removing the outer covers.
This is what it is supposed to look like when new, though:
And here is what the inside of mine looks like now:
Some cosmetic parts are going to be hard to get back into shape:
The way this furnace works, is that you place the part to be fired onto a pedastal and this is then moved into place using a motor. First, I checked that the motor still functions and is able to open and close the furnace, which checked out. Then I looked at the electronics and the brains of this operation is a Philips P80C552EBA
This particular micro does not have internal ROM, which means we should be able to get its program from an external chip. This chip is right next to it, an AM27C512 EPROM - made in 1986...
I temporarily extracted an Arduino Mega 2650 from a 3D printer - because I don't have an EPROM dumper and that is the only board I have on hand capable of the 5V levels necessary to drive this chip and sufficient number of pins. The program to extract the data is simple in principle, but even simpler in practice because someone else already wrote it for me :
To check that the data comes back properly, I read it twice. Let's look at the returned data in Meld:
Ugh! There is clearly some corruption in the ROM. Some data comes back differently every time I read it, but some data comes back consistently. I checked the connections, made sure all values from 0-255 are returned at least once. This proves that all the data lines are connected. I also made sure that there is not some sort of repetitive data coming back. That proves that all the address lines are connected. Lets check if at least some of the data is ok. First turn the text version of the hex data into an actual binary:
$ xxd -r -p VacumatHexMultiMultiread.txt > Vacumat.bin
And then open the result in a hex editor:
Ok, we can see the copyright string and date. This gives us some confidence that the whole process was carried out correctly and the ROM truly is corrupted.
I contacted VITA Gmbh via their facebook page, not expecting much help. They were quite friendly and asked me for my email address, which I gave to them, again expecting things to stop there. To my surprise I got a response from their "Head of VITA Furnace After Sales Department and Furnace Production"
thank you for your kind inquiry.
Herewith we would like to inform you that VITA does not provide single software programs, i.e. if required then you need to order the complete eprom through VITAs local destributor in your Country. The required spare part has got the part no. D30307.
Wow, that is pretty nice. I contacted the reseller for my region to see if they can provide me with this chip and how much it would cost. If I can get this, then I will try the disassembly one more time. If not, I guess I'll figure out what the original circuit board does and make my own.